For the InfiniiMax probes, what effect does the “Signal being probed” setting have? And my instrument’s firmware is A. If it fails even using the 2m cable, troubleshoot the test station. Message 10 of Customer should exercise caution when doing the hardware connection. Message 4 of 5. The current version of firmware can make gated measurements with a total time of ms.
|Date Added:||8 October 2018|
|File Size:||60.77 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Precision slotless female contacts are used to reduce the reflection uncertainty at a labbview interface. Obsolete View Service Options. What should I do to troubleshoot the instrument? Something so that the string you are sending to the device is not the one that is received.
Have you check on Agilents website? Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. My basic philosophy is that the data you are sending is not making it to the device for some reason.
What is the difference between a standard coaxial connector and precision slotless connector PSC? Message 9 of I would say that if this controller can talk to other devices, than the information that you are trying to generate is properly being generated. That is a terrible answer since I am not a big fan of “pointing fingers”, but to me that seems like the only logical conclusion.
I took a quick look at the user manual for the series and it appears that you should have no problem using this driver with the D.
query unterminated – Discussion Forums – National Instruments
They insisted that it cannot be a labvieew with the instrument – that it must be an error in MAX even though I was sending the simple command: Why do I have to specify single-ended or differential? Message 2 of 5. One is on a daisy chain with other instruments, the other was just connected directly to the USB port.
Message 8 of Message 7 of I used two different cables. I think this is highly unlikely. As you have probably already determined, the A and D versions have the same programming manual and aside from ,abview differences of the output range, they would appear to be compatible.
Also, I tried other instruments on the same cables and the communication was fine.
The UA power sensor is DC coupled. Message 5 of Slotless contacts make electrical connection with the mating male pin without changing the outer diameter of the female center conductor.
Do you get an answer in spite of the error? I have seen some instruments tht don’t do that and the cable could be then put on lzbview down.
Message 4 of 5. Message 2 of This has not always been the case, I have run several LabView programs on it. Legal Privacy Terms Trademark Acknowledgements.
When I send the reset command, it says invalid character. See search results instead: And my instrument’s firmware is A. Message 5 of 5. Main Page for instrument drivers: How do I execute?