I have successfully used several Generators Back to Top 2. I try to make this parameter point to the firmware path, but the program can’t be run. Take note that when the instrument is turned on, the GPIB Address that is assigned to the device is displayed. To help accelerate software development, all instrument drivers follow a consistent programming flow of:
|Date Added:||1 November 2013|
|File Size:||14.17 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Try it out sometime, because lsbview will come in handy later. Add an input to change the number of bursts you want the instrument to perform by right clicking the burst count terminal on the bottom side of the Configure Burst node and selecting Change to control.
Also notice then when you get into a submenu of the main menus, the blue arrow on the top left of the window becomes active.
EOL/obsolete: Keysight A signal generator
Back to Top 4. You can find these instrument drivers in the Instrument Driver Network. I used AgilentD scope to check AgilentA output directly via coaxial-cable. Also, my waveform is actually a bunch of pulses with various pulse widths, pulse to pulse intervals, and amplitudes hence the desire to define it myselfsuperimposed on a ramp function.
Jackie – Week 3. Consider PXI Arbitrary Function Generators NI PXI Waveform Generators can labvieew precise waveforms including sine, square, triangle, and ramp as well as arbitrary, user-defined waveforms by using data sequences or by streaming continuously from a host or peer-to-peer instrument within the PXI system.
Re-Introduction to LabVIEW
For this example, focus on the section between the Configure Arbitrary Waveform node and the Enable Output node. Place Controls onto the Panel using the Unplaced items box.
Once you get this part to work, look up some of the other commands in the manual on page Any information that appears in parenthesis after the name of the terminal is the default setting.
Using a simple resistive circuit set up the power supply and the multimeter to give different input labciew and measure the output across the resistor. To make a sinusoidal waveform use the command Sin 5.
Application Programming Interface API To help accelerate software development, all instrument drivers follow a consistent programming flow of: Right click terminals and choose Change to control to create a control wired to that terminal. If you use some clever programming you can get the program to respond just by changing a number here, or the push of a button there. Can any one please help me out to solve the issue.
How to use an binary stream file to generate Arb from Agilent 33250A?
Rate this document Select a Rating 1 – Poor 2 3 4 5 – Excellent. How to use an binary stream file to generate Arb from Agilent A? If I changed Freq, nothing changed of waveform. I have successfully labvkew several Generators You can configure the Configure Burst node with several inputs.
NI develops and supports thousands of instrument drivers for third party instruments to help engineers and scientists automate measurements. J’ai mis la valeur du dt egal la fruence d’hantillonage de ma boucle que je calcule.
Have your TA initial your sheet. How do I create such a file?
After above, I opened “ag33xxx – Getting Started 2. J’ai rssit faire se que je voulais avec “simulate signal”.
Automate Signal Generation From a Keysight/Agilent 33000 Function Generator With LabVIEW NXG
When writing an instrument control application for a serial instrument, you communicate with a COM port rather than the serial instrument connected to the COM port. Thanking You Robie 6. 33250 of these palettes is used in one window or the other. This tool automatically detects USB hardware connected to your host PC and adds a graphical representation to the window.
It will be a great help from your side.