Sign up or log in Sign up using Google. Hi Please describe which other instrument you are using. Message 1 of It can be DC or Sweep.. I’m not new to programming.
|Date Added:||8 July 2015|
|File Size:||61.20 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Which 64887 can be used to interchange between the two? My problem is that there seems to be a lack of documentation of drivers and I don’t feel comfortable with this going on with the project.
Keithley Picoammeter – Instrument Driver – National Instruments
I don’t know how to program this? Message 8 of Don’t follow the advice to install the keithley IO-layers.
Keithley 648X Series Meter
From your description, I would start with:. I am having difficulty in setting up the DC source and Resistance Sense of the picoammeter.
Most Active Hardware Boards: From your description, I would start with: However, before you start to mess around with any code, you should learn the instruments and how to perform a measurement manually. If you keuthley frustrated with labview and want to talk to it from a more ordinary general purpose language, the user’s manual for the instrument lists the corresponding SCPI commands for each manual operation.
Instrument Control (GPIB, Serial, VISA, IVI)
Sign up or log in Sign up using Google. Added is a small begin for a LabVIEW based instrumentdriver for the but this should be compatible I have limited knowledge about labview and have been trying to learn labview through online help and youtube videoes.
I will keitley the picoammeter to vary the voltage while simulataneously measuring the current. As far as my understanding goes, an IV curve can be obtained by sweeping the voltage across a predefined range and then measuring current.
Message 7 of Labivew program for KE picoammeter Solved! As is all programming in a new environment.
If you have mastered that, the rest is pretty easy. Thank you very much!
I think you have just saved me a lot of time and swearing! Keithey Resistance I am constantly getting an error that parameter 2 is out of range error in measurement function.
Keithley X Series Meter – IEEE (GPIB), Serial Driver for LabVIEW – National Instruments
Message 1 of This can be done by using a for loop, for instance. There is just some reference to how block should behave generally depending on the category they belong to such as Initialize, Configure, Data, ecc Message 6 of So here it goes. Icon and connector pane Description You won’t need the front panel, controls, or block diagram. Message 5 of I’m not new to keithlye.